Error1 is caused when part of the image area can not be analyzed. Potential causes may be: air bubbles in the sample, incorrect handling of the sample or abnormalities in sample.
Clear the error by;
- Take a new microcuvette.
- Repeat the measurement.
- Ensure you follow all the steps correctly and fill the blood completely in the microcuvette making sure there are no air bubbles.